GTM41134-WWVV-F Open Frame/Internal Medical Power Supply Datasheet
|Description||GTM41134-WWVV-F, Medical Power Supply, 60601-1-4th Ed. , Open Frame/Internal, Regulated Switchmode AC-DC Power Supply AC Adaptor, , Input Rating: 100-240V~, 50-60 Hz, Molex 09-65-2058 5 Position Header Pin 1: Earth, Pin 2: Removed, PIN 3: Neutral, Pin 4: removed, Pin 5: Line, Output Rating: 6 Watts, Power rating with convection cooling (W) , 3.3-48V in 0.1V increments, Approvals: UKCA; CB 62368; S-Mark 62368; LPS 60950; CE; EAC; WEEE; VCCI; China RoHS; RoHS; Ukraine; Book 60601; CB 60601-1 2MOPP; PSE; Level V; RCM;|
|CE EC-Declaration for Medical||https://globtek.secure.force.com/models/ec_declaration?id=a0Oa000000MEXnlEAH|
|CE EC-Declaration for Industrial/Commercial/ITE||https://globtek.secure.force.com/models/ec_declaration?id=a0Oa000000MEXnlEAH|
|CE EC-Declaration for Household||Contact GlobTek for more information|
|UKCA-Declaration for Medical||https://globtek.secure.force.com/models/ukca_declaration?id=a0Oa000000MEXnlEAH|
|UKCA-Declaration for Industrial/Commercial/ITE||https://globtek.secure.force.com/models/ukca_declaration?id=a0Oa000000MEXnlEAH|
|UKCA-Declaration for Household||Contact GlobTek for more information|
|Morocco SDoC Declaration||https://globtek.secure.force.com/models/Morocco_SDoC_Declaration?id=a0Oa000000MEXnlEAH|
|Prop 65 Statement||https://en.globtek.com/prop-65-statement.pdf|
|Conflict Minerals Declaration||https://www.globtek.com/pdf/conflict-minerals.pdf|
|Technology||Regulated Switchmode AC-DC Power Supply AC Adaptor|
|Category||Medical Power Supply|
|Input Voltage Range (V)||100-240V~, 50-60 Hz|
|Max Input Current (A)||0.6A Max|
|Max Output Power Rating (W)||6|
|Output Voltage Range (V)||3.3-48|
* (Final ordering #/Part No to be defined)
|A) ELECTRICAL SPECIFICATIONS:|
01. Input Voltage: Specified 90-264 Vac, Nameplate rated: 100-240Vac
02. IInput Frequency: Specified 47-63 Hz, Nameplate rated 50-60Hz
03. Output Current: Current limited to rated output current with tolerance of +/-10%
04. Output Regulation: +/- 5% measured at the output connector
05. Line Voltage Regulation: +/- 1% typical measures at full load
06. Output Ripple (Vp-p): 1% or 150 mV whichever is greater measured at 20 MHz bandwidth with 0.1 uf ceramic capacitor in parallel with 10 uf electrolytic capacitor connected at the end of output connector at nominal line
07. Turn-On/Turn-Off Overshoot: 5% maximum, 1mS typical recovery time for 25% step load
08. Turn-On Delay: 1 second typical
09. Hold-Up Time: 8mS typical @ nominal input voltage & full load
10. Inrush Current: 30A typical @ 115Vac input ; 60A typical @ 230Vac input
11. Switching Frequency: 65 KHz typical
1. Over-Voltage: Unit will recover upon removal of fault
2. Short Circuit: Electronically Protected, unit will recover upon removal of fault
3. Input Protection: Input line fusing
1. Dielectric Withstand Voltage: Input-Output: 5656Vdc, (2 MOPP)
2. Earth Leakage Current: N/A for Class II units, there is no PE Ground pin, so Earth Leakage current is not measured
3. Touch Current: Maximum allowed values: 100uA NC(Normal condition) 500uA SFC(single fault condition)
4. Means of Protection: 2 x MOPP
5. Primary to Secondary Bridging Capacitor: Standard configuration provided without bridging capacitors, Two (2) * Y1 type capacitors in series is available as an option upon customer request. If option employed, it is to be noted in the product deviation section of this specification document.
1. MTBF: 200,000 hours @ 25°C ambient temperature
2. Operating Temperature: 0°C to 40°C ambient temperature
3. Humidity: 0% to 95% relative humidity (RFP RFI-33542-1)
4. Storage Temperature: -20°C to 80°C
Normal performance after 2h recovery time at the room temperature without connection to mains (RFP RFI-33542-1)
1. Upper Housing: High impact plastic, 94V0 polycarbonate, non-vented
2. Lower Housing: High impact plastic, 94V0 polycarbonate, non-vented
3. Size: 43.5 x 74.0 x 35.3mm +/- 1.0
|Individual Box/Gift Box Size (mm)||90*45*30|
|Individual Box/Gift Box Color||White|
|Master Carton Size (mm)||455 x 405 x 220|
|Gross Weight (kg)||21.1|
|Label Width (mm)||25.4|
|Label Height (mm)||17.8|
|Description||Molex 09-65-2058 5 Position Header Pin 1: Earth, Pin 2: Removed, PIN 3: Neutral, Pin 4: removed, Pin 5: LineDocument: http://www.molex.com/pdm_docs/sd/009652058_sd.pdf|
Catalog Page: http://www.molex.com/molex/products/datasheet.jsp?part=active/0009652058_PCB_HEADERS.xml
|Popular Products (Stock items)|
An IEC 60601-1-2 medical grade rated power supply AC adapter converts input line voltage from the wall to a lower voltage, typically DC, which is used to operate and end use piece of medical equipment. IEC 60601-1-2 medical grade power supply AC adapters are typically housed in a plastic enclosure and are available with both 2 and 3 prong input options. power supplies are also available as open frame for use inside of medical equipment. These AC to DC converters are designed in accordance with international medical standards and provide for compliance with certain aspects of them which differs from generic information and communication technology equipment power supplies.
Why is my medical grade IEC 60601-1 power supply certified to IEC 60601-1 3rd Edition while the latest standard is IEC 60601-1 4th Edition?
IEC 60601-1 has different sections in specifications for safety and EMI/EMC. The latest revision of the safety portion of IEC 60601-1 is 3rd Edition while for EMI/EMC the latest revision is 60601-1-2, 4th Ed.
What is the effective date of 60601-1-2, 4th Ed?
60601-1-2, 4th Ed. was effective on December 31, 2018.
What's the difference between an IEC 60950-1 and IEC 60601-1 medical grade rated power supply?
IEC 60950-1 and IEC 60601-1-2 power supplies are both AC to DC converters and may be either internal or external but are designed to different specifications to match the end system specifications. While the basic performance behaviors are similar such as input and output voltages and mechanical configurations, there are many differences as well. Some of the key differences are that medical IEC 60601 medical grade AC to DC adapters are able to withstand a higher dielectric withstand voltage and must have lower leakage current. There are many other differences and also considerations to consider depending on the end use application.
What MOOP and MOPP?
Medical devices must incorporate one or more Means of Protection (MOPs) to isolate patients and operators from the risks of electrocution. A Means of Protection (MOPs) can be safety insulation, a protective earth, a defined creepage distance, or an air gap or other protective impedance. These can be used in various combinations. IEC 60601-1 3rd edition differentiates between the risk to patients and the risk to operators. A MOP can therefore be classified as Means of Patient Protection (MOPP) or a Means of Operator Protection (MOOP). The main difference between one MOOP and one MOPP is primarily one of permissible creepage distance. Both requirements are satisfied using Basic insulation. To achieve 2 x MOPP qualification the isolation test is particularly demanding at 4000 Vac and the creepage distance of 8mm is twice that required for one MOPP.
What is a BF and CF rated an IEC 60601-1-2 medical grade rated power supply AC adapter and what is the difference?
Medical grade power supplies and AC to DC adapters are categorized into one of three applied parts types: body type (B), body floating type (BF), and cardiac floating type (CF). These applied parts require specific attributes such as isolation voltage, creepage and leakage current. Floating refers to an ungrounded power supply, Class II, that is not referenced to another output and is fully isolated. Type B refers to devices that operate within the patient's vicinity, as described in IEC 60601-1 as six feet around a patient's chair or bed and do not touch the patient electronically and are grounded. These devices include operating room lighting, hospital carts and beds, automated dispensers of medication and supplies, and diagnostic equipment like x-ray machines. Type BF devices deliver electrical energy signals to or from the patient such as electrosurgical devices, endoscopes, and electrical vital sign monitors like thermometers. Type CF devices are similar to BF, except they supply power to devices connected to the heart, such as a defibrillator, external pacemaker or intracardiac ECG electrodes and Type CF requires a higher degree of protection against electrical shock than BF types.
What is an open frame or internal AC-DC Power supply?
Open frame AC-DC power supplies are also known as AC-DC converters and are a type of internal power supply. The open frame AC to DC power supply is built into the inside of a system and used to convert the AC current into DC voltage.
What is a IEC 62368-1:2018 power supply?
IEC 62368-1:2018 is applicable to the safety of electrical and electronic equipment within the field of audio, video, information and communication technology, and business and office machines with a rated voltage not exceeding 600 V. This is a product safety standard that classifies energy sources, prescribes safeguards against those energy sources, and provides guidance on the application of, and requirements for, those safeguards. The prescribed safeguards are intended to reduce the likelihood of pain, injury and, in the case of fire, property damage. An IEC 62368-1 rated AC to DC converter or power supply is a power supply that has been tested to and certified to comply with the said standard.
Did IEC 62368-1 power supplies replace IEC60950-1 power supplies?
On December 20, 2020 the safety standards that historically have applied to information technology equipment and audio video equipment, IEC 60065 and IEC 60950-1,transition to IEC 62368-1. In North America ETL and UL 60950-1 certifications will continue to stay valid and may continue to be used until further notice.
What is the difference between an IEC 62368-1 and a IEC 60950-1 power supply?
While the structure and contents of the standards are very different in general most products that previously complied with IEC 60950-1 comply with the latest standard IEC 62368-1 with little or no modification.
Will my IEC60950-1 power supply remain available?
GlobTek will continue to manufacture and supply IEC 60950-1 power supplies indefinitely until further notice.
How Does an AC-DC Adapter Work?
The AC to DC adapter or external power supply consists of a power supply which is enclosed in a housing similar to an AC plug, hence the term plug type, wall plug in, plug-in or plugtop power adapter. The power supply is plugged into the mains socket and the AC Voltage is then converted by the AC to DC adapter power supply into a DC voltage that is fed into the appliance via the output cable connection.